with all HP B accessories, and it can be added to any. HP B without affecting specifications. (An HP A must first be upgraded to an. HP B.). Checking the service manual, I found the circuit which managed it. It is an error factors and corrects the display for them, according the the HP Manual. Hp manual pdf: 0: General Discussion: Suncor. Manual for huskvarna chain sawa: 0: General Discussion. inwqmmjdn has reached a new hit record of
|Published (Last):||21 February 2005|
|PDF File Size:||12.93 Mb|
|ePub File Size:||18.40 Mb|
|Price:||Free* [*Free Regsitration Required]|
See each listing for international shipping options and costs.
There is a manusl to operate it without the RF unit, by placing a couple jumpers, one to start the unit and another one to select TEST mode.
This is the calibrator signal with several spans: As I suspected and John suggested, the extra gain was a problem. I began with the display section. So, all in all, it seems we are progressing: Format see all Format. I got a nice peak distortion so it went back to original position.
EB5AGV’s Workbench: HPA repair
So I have temporarily removed it and temperature has gone to reasonable levels, but still very hot. Once I got the unit back to the system, I got this nice display: Hp88566 I decided to get all cards out and carefully clean them and the card cage I am now reinserting them But here you have the tiny work done by the tech: So I cleaned and resoldered it carefully You will need to build bypass networks 4 of them.
In addition, there are separate bypass networks for the 3MHz filters, which you will need if you decide to adjust that in the future leave that for later, after the shack is remodeled. Here you have the display showing the fault: In that case, the manual calls for a Manual Calibration Procedureexplained also there.
Refine your search for hp service manual. Of course, I cleaned them. I was looking for bad capacitors, so this one seemed very suspicious look the discoloration in one side: Number of bids and bid amounts may be slightly out of date. Workmanship was not too fine so I redid it and also cleaned the PCB edge connector.
This narrow resolution bandwidth yields sensitivities to dBm. It seems gain is too much to be compensated by the auto-calibration system. If np8566 of the stages is not adjusted, the amplitude will be too high. Here you have a link to a very low quality video of it Condition see all Condition.
diagram HP B SPEC ANALYZER.
Excellent frequency stability, sensitivity, and frequency-reference accuracy combine to allow very accurate measurement of small signals in the presence of large manusl. Item Location see all Item Location.
All Listings filter applied. But, before moving it to its temporary storage location, I have checked it with the aid of my HPA generator another nice instrument!
EB5AGV’s Workbench: HP-8566A repair
The network consists of a resistor and a capacitor. More to come soon All in all, my main manjal for this SA will be in frequencies below 2. They seem fine enough, as there is none which exceeds 1dB: So I took my trusty ESR meter to check all of them: I have found them from 2.